## Support Center

### Setup, Recall, and Run Timed Program With Remote Commands for 1696, 1697, and 1698 Power Supplies

Last Updated: May 01, 2014 12:23PM PDT

In order to setup and recall stored programs saved into the internal memory for power supplies model 1696, 1697, and 1698, please look at the following command protocols and examples to describe how to do these actions via RS232 with remote commands.

PROM

location {1-9} Voltage {010-XXX} Current {001-

XXX}

Example: PROM003123456

This example sets V = 12.3V and I = 4.56A parameters and store to memory location 3.

This command sets the voltage and current to the location of the memory. You can assign from 1-9.

To recall the location, do this:

RUNM

location {1-9}

Example: RUNM003

This example will recall what was set above using the PROM command.

To set a timed program, use the command:

PROP

location {00-19} Voltage {010-XXX} Current {001-

XXX}Minute {00-99} Second {00-59}

Example: PROP00001234560216

This example will setup Step 00: V = 12.3V, I = 4.56A for 2mins 16secs

Repeat this command if you have multiple steps. So lets say you have another step, say Step 01. Change the location to 01 in the command like this:

PROP0001

After you setup the steps, you can run the program by # of cycles using the command:

RUNP

times {0000-9999}

Example: RUNP000020

This example will run the set program from above PROP command for 20 times.

Here's a complete example.  Suppose we want to do the following:

Step 00: V = 10.0V, I = 0.12A , 3min 48sec

Step 01: V = 6.5V, I = 7.89A , 1min 56sec

Step 02: V = 0.4V, I = 1.28A , 80min 2sec

Cycle Step 00-02 189x times.

We would send the following commands:

PROP00001000120348

PROP00010657890156

PROP00020041288002

RUNP000189

### Contact Us

178af1a33e8bb83a5155805675912147@bkprecision.desk-mail.com
https://cdn.desk.com/
false
desk
Loading
seconds ago
a minute ago
minutes ago
an hour ago
hours ago
a day ago
days ago
about
false
Invalid characters found
/customer/en/portal/articles/autocomplete