Applicable Models: 4075, 4078
Q: How do I accurately set the phase of two separate Arbitrary Waveform Generators to zero?
A: Display the outputs of the Sync connectors on an oscilloscope and adjust the phase until the leading edges of both signals are in sync. When this is done the phase of the two channels of your Arbitrary Function Generator or Arbitrary Waveform Generator will be in phase. You can also use a Phase Meter.